The Graphic User Interface (GUI) for your seneye is visable to any PC or Tablet on your LAN that has an HTML 5 supported browser. PC's or Tablets that are not on your LAN (ie connected by 3G) will not be visible.
The layout of the GUI is shown below.
The readings from the SUD are shown on an individual 'Tiles'.
If you hover your mouse pointer ther tile will rotate to show more options. There is a help link in the top right hand corner of the tiles that allow you to link to an answers page for more information
Each tile can be moved by clicking on the bottom bar and dragging it to a new location. The other tiles will automatically re-stack in a new location.
You can adjust the output for individual parameters from the SUD by clicking for a short time on the SUD settings tile. This will bring up the menu below.
This allows you to make minor adjustments for each individual reading by using the blue sliders. The reading + the trimed value is shown on the right hand side.
The 'Marine Tank' tick box (shown by the red arrow on the image above) needs to be set if you are using the seneye in a salt water aquarium. If you do not tick this, the pH reading will report high.
You can use the green ticks, shown on the right hand side of the display, to control the individual lights on the SUD. If you click on the tick, it will turn in to a cross and the light will be off all the time.
Click save, to complete the changes.